EPAX0500 Product Manual
DISPLAY: 4″ (101 mm) Red LED
5-Digit (EPAX0500): -19999 to 99999
EPAX Display: 85 to 250 VAC, 50/60 Hz, 10 VA
INPUT: Accepts analog input modules, see “Selecting Your Display Components and Option Cards.”
Display Indication: Three vertical dots on the left side of the unit identify the displays for the following modes:
Setpoint Indication: Four vertical dots on the right side of the unit identify the setpoint “ON” condition, with SP 1 being the top position through SP 4 at the bottom.
EPAX Programming: The unit is a large display, designed to be remotely mounted. Therefore, the unit does not have a programming keypad. Unit programming should be accomplished by one of the following methods: Rear Terminal Block: External switches can be wired via the terminal block to allow unit programming. A minimum of 3 switches would be required. Optional Programming Remote (EPAXPGM0): This option provides a 10 foot interconnecting cable and programming box. The Programming Remote contains buttons similar to the PAX, allowing easy programming of the EPAX display. Optional Serial Programming: Like all PAX units, you can purchase an RS232 or RS485 Comms Card and program the unit via Crimson, a Windows® based software program.
CERTIFICATIONS AND COMPLIANCES:
UL Recognized Component, File #E179259, UL61010A-1, CSA C22.2
Recognized to U.S. and Canadian requirements under the Component
Recognition Program of Underwriters Laboratories, Inc.
UL Listed, File #E137808, UL508, CSA C22.2 No. 14-M95
LISTED by Und. Lab. Inc. to U.S. and Canadian safety standards
Type 4X Indoor Enclosure rating (Face only), UL50
IECEE CB Scheme Test Certificate #US/8843A/UL
CB Scheme Test Report #04ME11209-20041018
Issued by Underwriters Laboratories, Inc.
IEC 61010-1, EN 61010-1: Safety requirements for electrical equipment
for measurement, control, and laboratory use, Part 1.
IP65 Enclosure rating (Face only), IEC 529
EMC specifications determined by the MPAX module.
Operating Temperature Range: Determined by the MPAX module
Storage Temperature Range: -40 to 60°C
Operating and Storage Humidity: 0 to 85% max. RH (non-condensing)
Altitude: Up to 2000 meters
Max. panel thickness is 0.375″ (9.5 mm)
Min. panel thickness for NEMA 4/IP65 sealing is 0.060″ (1.52 mm)
24-pin shrouded connector on EPAX engages connector on MPAX module
upon installation. Shroud ensures proper alignment by providing a lead-in for
the module connector.
10. CONNECTIONS: Wiring connections are made to the EPAX terminal
block and MPAX module via high compression cage-clamp terminal blocks.
MPAX Module Wiring: Instructions are provided in the corresponding PAX
EPAX Terminal Block Wiring:
Wire Strip Length: 0.3″ (7.5 mm)
Wire Gage: 30-12 AWG copper wire
Maximum Torque: 5-7 inch-lbs (0.58-0.81 N-m)
CAUTION: DISCONNECT ALL POWER BEFORE
INSTALLING OR REMOVING MODULE
CONSTRUCTION: Aluminum front panel, enclosure, and rear cover with
textured black polyurethane paint for scratch and corrosion resistance
protection. Sealed front panel meets NEMA 4X/IP65 specifications for
indoor use when properly installed. Installation Category II, Pollution Degree
2. Panel gasket and keps nuts included.
WEIGHT: 5 lbs (2.25 kg) (less module)